ACCUTREND PLUS MANUAL PDF
to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||18 August 2009|
|PDF File Size:||4.18 Mb|
|ePub File Size:||1.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
Performing a measurement Switching the code display You can switch between the display of the stored code numbers for information purposes. Press the Set button to afcutrend the displayed year.
Cobas Accutrend Plus User Manual
This starts the actual measurement. Quality control recommendation Open the measurement chamber flap. Press the M button to change the currently flashing figure.
Important Notes Performing a wccutrend Important notes Always … operate the instrument at the acceptable test-specific temperatures see also the test strip inserts: The Accutrend Plus instrument Power To save power, the instrument powers itself off after mnaual minutes unless a button is pressed or a new test strip is inserted.
The instrument is suitable for professional use as well as for self-testing. Instrument settings The date and time as well maunal the set symbol now appear on the display. Performing a measurement A flashing arrow now instructs you to open the measurement chamber flap to apply the blood.
Page 35 Instrument settings If the displayed setting is correct e. Every time the instrument is powered on, you can check the display. Product Specifications Further information Product specifications Operating conditions and technical data Temperature range for Cholesterol: Page 10 Contents Product specifications Operating conditions and technical data Setting The Date Instrument settings Setting the date With the next three settings, you first enter the year, then the month and finally the day.
When the battery symbol is shown with a measurement screen except during the display testyou will only be able to perform a few more measurements.
Target values which should be obtained can either be found manua the labels or on the package inserts for the test strips or control solutions. Quality control recommendation Display of results When measurement is finished, the result is displayed. Lift the lid upwards to remove it from the instrument.
If the problem persists, please contact your local customer accutrrend and service center. Accuternd a measurement The flashing drop symbol above the strip symbol now instructs you to apply the blood. Page 93 Memory After the usual display test has been performed when switching on the instrument, the last stored measured value is displayed. Corrections can only be made by repeating the settings. Page 95 Memory To navigate within one memory area, use the M button.
You can flag a measured value at different points in time: The batteries are potentially hazardous also and should be disposed of accordingly. Cleaning If needed, you can rinse the strip guide accuyrend from the instrument under warm running water. Temperature The instrument or ambient temperature does not fall within the acceptable range for the selected test.
Page 81 Quality control recommendation What you need: Press the Set button to save the setting. Manaul properties of these test strips are then stored in the instrument.
Performing a measurement Check that the whole display is functioning correctly to prevent misinterpretations due to defective display elements.
Alphabetical Index Alphabetical index Alphabetical index Capillary blood At the start of the measurement after inserting the test strip. Cleaning Cleaning the optical measuring system Clean the easily accessible areas of the optical measuring system under the strip guide with a lint-free pad, a moistened cotton ball, or a moistened cotton swab. Press the Set button to switch to the code number for the other parameter.
Alphabetical index Operating the instrument The display then automatically switches to the mode for setting the time format. Moving backwards is not possible. Refer to the insert of the respective test strip.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
The code strip accutfend required at least when a new test strip container is opened and before the strips are used.
Operating the instrument Operating the instrument Before using the instrument for the plu time, perform the following steps: Performing a measurement Flagging measurements You can add additional information to measurements, for example, to characterize a particular value with regard to special circumstances e. Brief Overview Of The Steps To Be Carried Out Performing a measurement Brief overview of the steps to be carried out Code the instrument once Insert the test strip Apply blood per test strip lot Page 46 Performing a measurement Start measurement by closing the Display of measurement results measurement chamber flap When the test strip reaches the correct position, you will hear two beeps short — Introduction Integrated control functions The Accutrend Plus instrument has numerous integrated or available afcutrend functions, including the following: Instrument settings Setting the time First enter the current hour and then the minutes.
Medical staff and other people using the Accutrend Plus instrument to perform measurements for more than one patient must be aware that any object coming into contact with human blood accutrsnd a potential source of infection. Performing a measurement If the displayed result particularly when blood glucose is measured does not match your state of health or seems accutrejd be unusually high or low, check the meter function using a new test strip see the Control Check with Control Solution chapter of this manual.
If a single measured value is stored incorrectly, one dash — will appear in the display. Instrument settings General procedure for setting up the instrument set mode You make all settings using the Set and M buttons as described below. Page Alphabetical index Date Operating the instrument Instrument settings Brief overview of the instrument settings The following table provides an plu of the available settings.